≤DP128-A-01-040 Diffusion Phantom ADC Values
Measured at NIST Boulder at 3.0 T, 0°C.
Data is relevant to Diffusion Phantoms with serial numbers ≤DP128-A-01-040.
Data measured and published by NIST Boulder, the Magnetic Imaging Group (Keenan, K. E., et al. “Assessing changes in MRI measurands incurred in a scanner upgrade: Is my study compromised?.” Proceedings of the International Society for Magnetic Resonance in Medicine Scientific Meeting and Exhibition. Vol. 25. Honolulu Hawai’i, 2017.)
Last updated December 14, 2021, 10:00 PM MST
≥DP128-A-01-041 Diffusion Phantom ADC, T1, T2 Values
Measured at NIST Boulder at 3.0 T at 0°C and between 16°C and 26°C.
Data is relevant to Diffusion Phantoms with serial numbers ≥DP128-A-01-041.
Data measured and published by NIST Boulder, the Magnetic Imaging Group (Report of Measurement: Magnetic Resonance Measurements for MRI Biomarkers: Water Diffusion Coefficient, NIST Boulder, Certificate Number TBA).
This report is a DRAFT. It will be made official very soon.
Last updated July 21, 2021, 6:00 PM MST